Validation of indwelling vaginal sensor to monitor body temperature in ewes

Remote-sensing technology facilitates longitudinal collection of body temperature during periods of hot or cold environmental stress without human interference, producing high-frequency measurements whilst reducing labour and stress to the animal. A pilot study was conducted to validate an improved, minimally invasive method for the continual measurement of vaginal temperature (VT) in female sheep. A silicon mould of a modified controlled internal drug release (CIDR) device (Zoetis Animal Health, Parsippany), was manufactured to securely house a temperature logger (Micro-T 16-bit; Star Oddi, Iceland) and allow direct contact at the collection site. These temperature-sensing CIDR devices were validated against manual rectal temperature (RT) measurements collected from 15 mature, nonpregnant, non-lactating Merino ewes. Rectal temperature was measured from each individual, six times per day in 2-h intervals for 14 consecutive days. The simultaneous measures of VT and RT did not differ significantly within each ewe (P > 0.05) and demonstrated a moderate linear relationship (R = 0.62, P < 0.05). The mean (± SEM) difference between RT and VT was small (0.010 ± 0.004 °C), with a 95% confidence interval of − 0.26 to 0.29 °C. Additionally, the coefficient of variation was lower on average for VT (0.49%) compared to RT (0.59%). Differences among paired readings were likely due to interval variation as well as penetration depth, air influx and faecal temperature when collecting RT. The initial silicone manufacturing costs were high (2200 AUD), with each unit costing 1.25 (AUD) thereafter; however, the temperature-sensing CIDR device was an effective and efficient research tool for the remote monitoring of body temperature. While further validation of these devices within extensive grazing environments is warranted, the collection and analysis of longitudinal physiological data from ewes has the potential to improve a variety of management aspects related to extensively grazed ewes maintained and joined in harsh conditions.


Introduction
Climate change is the most serious environmental challenge faced by producers around the world, threatening the physiology, reproductive function and overall health and productivity of livestock [1]. Heat stress often occurs when the effective environmental temperature exceeds the upper critical temperature of a sheep (25 to 31 °C) [2], resulting in an imbalance between heat production within the body and its dissipation. Biologically, animals have the ability to minimise the negative effects of hot conditions by invoking physiological mechanisms [3], such as an increase in respiration rate and body temperature, as well as by adjusting their activity and feeding patterns to avoid hot periods of the day or increase the use of shaded areas [4]. An increase in core temperature is one of the primary physiological indicators and has been used to evaluate the heat stress response in sheep [5]. It is Lewis Baida et al. Animal Biotelemetry (2022) 10:6 also related to a number of other varied functions such as activity, nutrition, health preservation and reproductive success [6,7]. Manual measurement of temperature using a digital rectal thermometer is considered the best estimate of core body temperature and is currently the most reliable diagnostic tool for identifying illness [8], disease [9] or heat stress [5] in livestock animals. Rectal temperature (RT) is relatively accurate and repeatable; however, the presence of humans can often disrupt the behavioural and physiological responses of an animal and lead to stress-induced hyperthermia [10]. Additionally, manual thermometry is labour intensive and only provides a cross-sectional sample due to the need to handle and restrain the animal. For this reason, manual thermometry is not suitable for the longitudinal assessment of body temperature patterns in free-ranging animals [11]. To circumvent the issues associated with manual collection, a plethora of automated technologies are being trialled for the continuous, remote measurement of body temperature in livestock animals [12].
Technologies appropriate for the automated collection of heat stress data in sheep have been recently reviewed [13]. Many tools used to measure core temperature, such as subcutaneous microchips or rumen boluses, require surgical intervention or are otherwise invasive [14,15]. These methods are also not always practical in a research setting due to their irretrievability or fatal outcome for the animals, and orally administered rumen sensors are affected by internal environmental changes, such as drinking events [16]. A non-invasive, contactless approach is the use of thermal imaging or infrared thermography (IRT). Despite being a relatively useful tool for monitoring body temperature in sheep [17], skin or peripheral temperature can vary depending on the presence or absence of wool and readings can be affected by ambient temperature and UV light, leading to inconsistencies at different anatomical locations [18,19]. Another less invasive and more practical method is the use of indwelling temperature probes [20][21][22]. Despite consistently recording core temperature in livestock species, rectal probes require external physical attachment to secure the device and resist expulsion during defecation [13,19]. Although this has been achieved in cattle with the use of a tail harness [23], a reliable attachment method has yet to be developed in sheep, which typically lack a comparable tail. Difficulties in keeping the temperature probe stationary as well as expulsion and faecal temperature during defecation therefore limit the practicality and reliability of this method in sheep. These limitations have led to the collection of vaginal temperature (VT) and the use of indwelling vaginal probes in female sheep [24], as differences in VT and RT have been found to be negligible when using identical temperature measuring devices [25]. Affordability, practicality and welfare concerns can affect how body temperature is measured, including the frequency of measurement as well as the type of device which is used. In addition to comparing the measurement of body temperature at different anatomical locations, different types of temperature loggers have also been compared by measuring VT with two sensor types simultaneously within the same animal [26].
The most common deployment technique for inserting temperature loggers into the vagina is the use of controlled internal drug release (CIDR) devices. Usually containing progesterone to synchronise oestrous, these devices are instead sourced as blanks (i.e. no hormonal treatment) and used to house a self-contained temperature logger. The same technique has been used in both sheep [21] and cattle [27], whereby a segment of the CIDR was removed, replaced with a temperature-sensing device and secured in place with tape or shrink tubing. Despite previous studies reporting the relative effectiveness of this method [21,22,27], temperature sensor coatings such as tape and/or shrink tubing can influence body temperature measurements. As most temperature sensors used to study thermoregulation are coated with a suitable material to waterproof the device as well as to protect the animal from infection [28], extra coatings applied to secure the device in place can influence accuracy and response time for core temperature data collection. Due to huge variability in tag design, deployment methods and animal physiology, it is not possible to make a general quantification on the effect of sensor coatings on body temperature measurements. However, the materials used to coat the devices are known to thermally insulate the sensor, potentially biasing temperature measurements and indicating an improved thermoregulatory response than what actually exists [28]. Here, we present an alternative method for deploying an indwelling vaginal temperature sensor in Merino ewes in order to eliminate the risk of bias associated with additional material layers or coatings. This newly engineered deployment design allows direct contact between the temperature sensor and the skin inside the vagina, giving the sensor the ability to accurately capture acute fluctuations in core temperature. As a result, this study successfully validated an improved deployment method by evaluating the relationship between manual rectal temperature and automated vaginal temperature in Merino ewes.

Animals
This was a collaborative experiment between The University of Adelaide and South Australian Research and Development Institute (SARDI), and conducted in latesummer, February 2021 at the Turretfield Research Centre, Rosedale, South Australia (34° 33ʹ S, 138° 50ʹ E). All animal use, housing and methodology were approved by the PIRSA Animal Ethics Committee (project: 11/12) and conducted in accordance with the provisions of the Australian Code for the Care and Use of Animals for Scientific Purposes.
Both RT and VT data were collected from fifteen 4-year-old South Australian dry (non-pregnant, nonlactating) Merino ewes (Ovis aries) (75.8 ± 1.6 kg body weight). VT was collected using a newly developed indwelling vaginal probe, whilst RT was collected manually using a high-accuracy digital thermometer. Ewes were randomly selected, moved from the paddock and split into two outdoor yards (10 m 2 ; pen 1, n = 8; pen 2, n = 7), for a 6-day acclimatisation period. Ewes were then moved to an indoor facility and randomly housed in individual pens (2.5 × 1.5 m) for the remainder of the study (14 days). VT sensors were inserted into the ewes upon moving into the indoor facility and ewe temperature data were collected at 10-min intervals. Restraint of ewes by experienced personnel in a pen or raceway was required to instal the device, taking less than 1 min per ewe. As risk of irritation or infection increases as the length of deployment time increases, each ewe was inspected several times each day for irritation and/or device dislodgement. Signs of irritation/infection include excessive discharge, increased temperature and inappetence [29].
Between 0800 and 1800 h and 2 h apart, six RT data were manually collected, with exclusions at 0800 h on day 1 and 1400 h on day 8 which were accidentally missed and no data recorded. RT was collected by two trained personnel inserting a high-accuracy digital thermometer (One Step ® ) approximately 5 cm into the rectum and pressing against the wall of the rectum until a stable reading was reached (~ 30 s). Ambient temperature and relative humidity within the indoor housing facility was measured using a HOBO Temperature/Relative Humidity Data Logger (MX2302A; Onset Computer Corporation, USA), preprogramed to collect measurements every 10 min. During the study, ewes had ad libitum access to water and were fed 500 g of grain/ewe/day as well as rations of hay to ensure their energy requirement for maintenance was achieved.

Device construction
Modified CIDR devices and applicators were reverse engineered and moulded (AZ3D Pty Ltd.) using a combination of two manufacturing techniques: computer numerical control (CNC) machining and hydraulic press manufacturing. Initially, a P20 steel single cavity mould was created using CNC machining. Thereafter, hydraulic press manufacturing was used to produce 200 probes in a biocompatible silicon material with a shore hardness of 70. Design modifications were made to an original sheep CIDR device (Zoetis Animal Health, Parsippany, NJ), to enable the secure housing of a Star-Oddi Data Storage Tag (DST) (Micro-T 16-bit; Star-Oddi, Iceland) ( Fig. 1). Modifications included adding curvature to the sharp edges of the probes to ensure the data logger was securely held in the cavity, along with increasing the diameter of the tail of the probe. Manufacturing cost of the silicone mould was 2200 AUD and following this, cost per unit was 1.25 AUD. The temperature loggers were placed inside the centre of each of the CIDR moulds, exposing the cylindrical sides of the logger and enabling direct contact between the logger and the skin inside the vagina (Fig. 1). These temperature sensors had an accuracy of ± 0.06 °C and were factory-calibrated and tested prior to receipt from the manufacturer. All loggers had a threepoint traceable calibration certificate at 10 °C, 25 °C and 40 °C, specifically calibrated for use at 5-45 °C. The logger was pre-programmed through a communication box and Mercury software (Star Oddi, Iceland). Devices were inserted approximately 10 cm into the vagina of the ewes using a modified CIDR sheep applicator (Zoetis Animal Health, Parsippany, NJ) (Fig. 2), after applying Savlon ® antiseptic cream to the tip of the applicator. At conclusion of the experiment, the devices were removed and data were downloaded with the communication box, exported to Microsoft Excel v. 2016 (Microsoft Corporation, Redmond, WA); and labelled with device/logger ID, ewe ID, date and time.

Statistical analysis
Statistical analysis was conducted using both IBM SPSS Statistics v. 26 and Xrealstats Statistics Package for Microsoft Excel v. 2016 (Microsoft Corporation, Redmond, WA). We excluded all VT data collected up to 1 h after insertion, as they were likely influenced by the insertion process. In order to compare VT and RT, we matched them with reference to animal ID and time-point. The manual RT data collection usually took 20 min and therefore, the mean VT based on the VT readings captured across this 20-min period was calculated. Automated VT and corresponding RT data were analysed using a linear regression to calculate the coefficient of determination. As the best anatomical location or methodology for determining an animals' true core temperature has yet to be defined, both VT and RT were estimated measures of core body temperature, and a relationship between these two measures was expected. As the coefficient of determination does not allow to determine the difference between measures, we also investigated the agreement between automated VT and manual RT using the Bland-Altman method [30]. We determined the strength of agreement, direction, and upper and lower agreement between VT and RT. To determine the concordance of VT against RT, a Pearson's and Lin's concordance correlations coefficients (LCCC) [31] were also conducted. Data are presented as means ± standard error of the mean (SEM).

Results and discussion
The use of indwelling temperature sensors allowed for the automated measurement of VT at 10-min intervals in the absence of a human observer. The temperature sensors (DST Micro-T 16-bit; Star Oddi, Iceland), contain two centrally positioned thermistors and are coated in a highly thermally conductive ceramic material, which facilitated fast and uniform heat distribution throughout the whole device. Although relatively expensive, these loggers have the advantage over cheaper alternatives due to their high accuracy (± 0.06 °C), sensitivity and rapid response time [26]. The replicas designed for this experiment eliminate the need for extra material to secure the loggers (shrink tubing, vinyl electrical tape, rubber, etc.), which has shown to create inconsistencies and decrease device sensitivity [28].
Considering the data logger as the experimental unit, n = 82 paired observations per logger were generated and used to validate the temperature loggers in vivo. Mean (± SEM) VT and RT at each time point were compared across the 14-day trial period to determine whether a similar pattern existed (Fig. 3). VT profiles of all animals remained relatively stable outside the manual data collection period; however, acute fluctuations in VT were consistently observed between 0800 and 1800 h each day which occurred during the period of manual RT collection. Although a precise explanation for these acute fluctuations cannot be elucidated, it is likely they were induced by human presence during manual data collection. Although domesticated, sheep are a prey animal and Fig. 2 A CIDR replica inserted into CIDR applicator and, B CIDR replica and logger ejected from applicator the behavioural responses of their wild ancestors evolved to evade detection and capture by predators [32]. For this reason, exposure to a stressor such as human handling can activate an endocrine response, leading to stressinduced hyperthermia (SIH) and increased metabolic heat production associated with the flight response [10]. Unlike studies which make direct comparisons between anatomical locations using the same, or varied automated technologies [25,27,33], the current study compares the use of automated technologies versus manual measurement. Despite some using a similar approach in cattle [20,34] and sheep [15,20], researchers are yet to report how body temperature is influenced by the collection of RT. Following the manual collection of rectal temperature by researchers, an increase in mean ewe VT between 0.2 and 0.6 °C was observed at each collection time-point on each day, with the maximum increase recorded at 1800 h on day 10 of the trial period. Increases in VT were consistently seen to reach a peak at approximately 30 min after manual RT collection and before decreasing back to basal range. SIH in sheep is a common response to emotional or physiological stress, often caused by routine husbandry practices such as shearing [35] and transport [36]. When shearing was used to stimulate SIH, it was found to last between 4 and 14 min post-shearing when automated body temperature was taken in 1-min intervals [35]. A higher collection interval in the current study would have allowed a more accurate measurement of the duration of SIH caused by human handling/restraint. Although diurnal variation across individual sheep is normal [37], acute core temperature changes of ≥ 0.5 °C are considered a physiological response and in some cases, a welfare concern [38]. Increases in VT due to SIH seen here, are therefore biologically relevant and would introduce a level of interference if manual RT was used to quantify heat stress based on elevated core temperature. Hence, studies which have previously used manual RT as an indication of heat stress [39], may have biased results due to the influence of human interference and occurrence of SIH. Feeding occurred prior to data collection at both the 0800 h and 1800 h time points, which provides reasoning for a larger and extended increase in VT due to prolonged human presence at these time points. More research is necessary to truly define the influence of manual thermometry on core temperature fluctuations in sheep. Nevertheless, comparing each method simultaneously allows us to understand the relationship between varying techniques as well as anatomical localities.
The coefficient of determination indicated that there was a moderate linear relationship between VT and RT collected at the same time points (R 2 = 0.62, P < 0.0001; Fig. 4). The coefficient of variation was lower on average for VT (0.49%) compared to RT (0.59%), suggesting that there was less variation in VT in the current study; however, lower VT variation may be due to VT being averaged over three measures during the manual RT collection period. A previous study has also suggested that a strong relationship exists between manual RT and automated VT in lactating ewes after administration of lipopolysaccharide (R = 0.97, P < 0.001; [20]), as well as in lactating control ewes within the same experiment (R = 0.88, P < 0.001; [20]), and in gestating Senepol cows (R = 0.78, P < 0.001; [20]). Similarly, Vickers et al. [34] investigated the same relationship in dairy cows postpartum (R = 0.81, P < 0.001); however the strength of the relationship decreased by almost half during peak lactation (R = 0.46, P < 0.001). The relationship identified within the current study (R 2 = 0.62, P < 0.0001) was lower than that previously described in sheep (R = 0.88, P < 0.001; [20]); however, different data collection and statistical methods were performed, making direct comparisons between the two studies difficult. Studies which compare the concurrent measurement of RT and VT using only automated temperature sensors tend to improve the relationship observed between the two anatomical locations (R = 0.92, P < 0.0001; [27]). Given that the current study used a combination of indwelling data loggers and a high accuracy hand-held thermometer to obtain time-point measurements, intervals were slightly variable. Additionally, when taking manual RT measurements, variations in penetration depth, air influx and faecal temperature could also contribute to differences encountered between RT and VT [34]. It is also important to consider the changes in uterine and vaginal blood flow which occurs at varying reproductive stages, which has been reported to alter VT in sheep [24,40]. Regardless, the current study is the first to evaluate the relationship between VT and RT in dry Merino ewes using a novel deployment technique. Calculating the correlation coefficient (R) or coefficient of determination (R 2 ) describes; (a) the relationship and direction between variables or, (b) the proportion of variance that two variables have in common, respectively [41]. Considering that RT and VT are both estimated measures of core body temperature, a close relationship between the two measures was anticipated. Therefore, results generated from the above models are sometimes inadequate and may be misleading, as a high correlation or linear relationship does not imply that there is strong agreement between two methods. An alternative method was introduced by Bland and Altman [30], which is used to quantify the agreement between two quantitative measurements by constructing limits of agreement [41]. Analysing RT and VT using the Bland-Altman approach [30,42], ultimately determines whether two methods of measurement are comparable. As the body temperature of sheep is maintained within a tight range, 38.3-39.9 °C [43], the Bland-Altman method of comparison was performed by using RT minus VT. In the current study, this model indicated that the mean difference between RT and VT was small (0.010 ± 0.004 °C), with a 95% confidence interval of − 0.26 to 0.29 °C (Fig. 5). Although the coefficient of variations suggests that automated measures of VT may be a more precise estimate of core body temperature in sheep, it is important to consider that VT was averaged over three measures during the manual RT collection period. Nevertheless, these results suggest that RT and VT are comparable and reproducible.
Secondary to the Bland-Altman method of comparison, measuring the reproducibility of a method from trial to trial is also of interest. For this purpose, an index developed by Lin [31] which evaluates agreement between paired readings has advantages over other correlation models as it considers each measurement as a distinct reading rather than a replicate (random). The LCCC for the current study was 0.78. Although there is little agreement and differences in stringency as to how to interpret LCCC values [44], values ≥ 1.0 indicate strong concordance between two measures. Below this, LCCC is often interpreted similar to other correlation coefficients like Pearson's, where < 0.2 is poor and > 0.8 is excellent [45]. Others have a more rigorous interpretation, classing LCCC values < 0.9 as poor [46]. Reasoning for such a high threshold is likely because LCCC usually gives a higher value for most practical situations [47]; however, this is often too high to be reasonable. The moderate LCCC value of 0.78 in this study is in agreement with the coefficient of determination (R 2 = 0.62) and suggests a lowmoderate concordance or reproducibility of the methods, in alignment with the approach made by Altman [45] and others [48].
The body temperatures of mammalian species exhibit ultradian and circadian rhythms around a set point [49], which has shown to be influenced by climate [37].
When assessing the circadian rhythm in ruminants, body temperature is at its lowest in the morning and highest in the evening [50]. If body temperature is to be used to infer heat stress in free-ranging sheep, their daily temperature patterns need to be well understood. It is difficult to establish a definite circadian rhythm in the current study due to the interference during manual data collection. Nevertheless, a gradual decrease in vaginal temperature was consistently observed between 1900 and 0700 h the following day, with the highest temperatures occurring at approximately 1800 h (Fig. 3). In the absence of a human observer, the indwelling data loggers validated in this study would have the capacity to accurately assess the natural thermal rhythms in free-ranging sheep compared to manual collection using a hand-held thermometer. Similarly, due to the high-sensitivity of this technology, it has the potential to determine varying stages of the reproductive cycle in livestock. Changes in uterine and vaginal blood flow which occur at different stages of reproduction [40], have been shown to alter the VT of sheep during gestation as well increasing it during peak estrous [24]. Under consistent or controlled ambient conditions, these vaginal loggers have the capacity to Mean differences (dashed line) and confidence intervals (95% = mean ± 1.96 × SD; solid lines) are also indicated. The x-axis represents the mean temperature measurement as determined by averaging rectal temperature (RT, °C) and vaginal temperature (VT, °C), whilst the y-axis shows the difference in recorded temperatures for the two methods, in this instance RT-VT better define how varying reproductive stages or hormonal activities throughout estrous can influence VT in ewes.
Weather conditions at the time of the experiment were relatively stable across the trial period. Although the experiment was conducted indoors, the facility was not temperature controlled and the sheep were therefore exposed to ambient conditions similar to those outside the housing facility. Ambient temperature measured inside the housing facility did not vary more than 2 °C from that recorded by the nearest, on-farm weather station (Gazeeka WeatherBox, Vomax Instrumentation Pty Ltd ©, Marleston SA). Despite ambient temperatures being naturally high during the trial period (Table 1), most sheep were able to maintain their core temperature below their upper limit (39.9 °C), as indicated by the automated collection of vaginal temperature (Fig. 6).
Despite all data being downloaded at the conclusion of the study with 0% data loss, there is a potential for data loggers to either become dislodged or fail to collect and store data successfully, which can only be recognised during the data download period. Another limitation of indwelling loggers is that prolonged use may lead to infection or expulsion. Fortunately, the current study found little to no irritation or damage posed to vaginal tissue. VT measurement produces a reading that more closely reflects core temperature and is the most consistent with RT readings when compared to peripheral and subcutaneous sensors [22]. However, the device is a data logger not a radiotelemetry device, meaning that data are stored and must be downloaded. Radiotelemetry technology would be especially useful in an industry setting, as all data would be communicated and transcribed to a database with the ability to monitor body temperature in real-time. Nevertheless, automated data loggers are a reliable and advantageous method of obtaining accurate body temperature measurements within a research setting.  [43]. Individual ewe variance for VT ranged from 0.029 to 0.056 °C

Conclusions
In the current study, VT and RT were found to demonstrate a moderate linear relationship, level of agreement and reproducibility. The improved deployment technique trialled in this study facilitated the accurate measurement of VT and proved to be a reliable method for the automated collection of core body temperature in dry Merino ewes. Although VT can only be measured in female animals, it is minimally invasive and considered the best site for the short-term measurement of body temperature in female sheep. A major finding in this study was the marked effect that human interference has on acute fluctuations in body temperature. Stress-induced hyperthermia is an important evolutionary mechanism and prey animals must be aware of their environment and respond properly to all kinds of stimuli. This highlights the importance of decreasing the need for human handling and restraint during the collection of physiological data, such as body temperature. Due to the ability to take continuous measurements over time without the need for a human observer, indwelling data loggers are undoubtedly more useful in a research setting and for evaluating heat stress in sheep. Future research using this automated approach, will have the ability to accurately monitor the body temperature of free-ranging ewes and quantify the impact of hot conditions on the ability of ewes to maintain their core temperature within normal range. The temperature loggers used in this study have enough sensitivity and accuracy to define the thermal thresholds of individual sheep. Identifying differences in the thermal thresholds of individual ewes could then allow investigation into the transgenerational effects of heat tolerance.